NI USB 6009 WIRING DIAGRAM
NI USB-6008/6009 User Guide - National Instruments
PDF fileThe National Instruments USB-6008/6009 devices provide eight single-ended analog input (AI) channels, two analog output (AO) channels, 12 DIO channels, and a 32-bit counter with a full-speed USB interface. This user guide describes how to use these devices. For specifications, refer to the NI USB-6008 Device Specifications and the NI USB-6009
Archived: Interactive Control Panel for USB-6501, USB-6008
NI-DAQmx Base offers a subset of NI-DAQmx functionality on Windows, Linux, Mac OS X, Windows Mobile, and Windows CE OSs. The Interactive Control Panel is an example program that ships with the NI-DAQmx Base driver software and provides users with a quick-start way to begin using their new USB-6501, USB-6008, or USB-6009 hardware.
NI USB-6008/6009 User Guide and Specifications
PDF fileNI USB-6008/6009 User Guide and Specifications 2 ni Figure 1 shows key functional components of the NI USB-6008/6009. Figure 1. NI USB-6008/6009 Block Diagram
My USB-6008/USB-6009 Ouputs Incorrect Voltage Readings - NI
Another thing to note is that the pinouts for the NI USB 6008/6009 OEM cards are not the same as the pinouts for the non-OEM version. If you are using the OEM version, the correct pin diagrams (shown in "Additional Information") are found in the NI USB 6008/6009 OEM User Guide.
National Instruments NI USB-6008-6009 User Manual
The NI USB-6008/6009 creates an external reference and supplies a power source. All voltages are relative to COM unless otherwise noted. +2.5 External References. The NI USB-6008/6009 creates a high-purity reference voltage supply for the ADC using a multi-state regulator, amplifier, and filter circuit.
USER GUIDE AND SPECIFICATIONS USB-6008/6009
PDF fileThe NI USB-6008/6009 provides connection to eight analog input (AI) channels, two analog output The following block diagram shows key functional components of the USB-6008/6009. Figure 3. Device Block Diagram as indicated by the overlay label on the USB-6008/6009 device. 3. Connect the wiring to the appropriate screw terminals.
How to connect a mechanical switch to the USB-6009 - NI
How to connect a mechanical switch to the USB-6009. How to connect a mechanical switch to the USB-6009 Flasher. Member 05-25-2011 Please see attachment for wiring diagram. If the program is not right or my connection is wrong, please help me to verify the problem
USB-6008/6009 - ULisboa
PDF fileUSB-6008/6009 User Guide 6 ni Hardware The following block diagram shows key functional components of the USB-6008/6009. Figure 4. Device Block Diagram
NATIONAL INSTRUMENTS 6008 USER MANUAL AND
View and Download National Instruments 6008 user manual and specifications online. Page 1 USER GUIDE AND SPECIFICATIONS NI USB-6008/6009 This guide describes how to use the National Instruments USB-6008/6009 data acquisition (DAQ) Signal Label Application Diagram Connect the wiring to the appropriate screw terminals.
Connecting 2, 3, and 4 Wire RTDs to My Data - NI
It is not recommended to use a USB-6008, USB-6009, or any M, E, or S series cards for temperature measurement because RTD's require a current excitation source. These devices will not be able to provide this current excitation, so you will need an external supply.